What are the important safety precautions?
- Remove anything conductive from your hands (e.g. metal rings, watches, or bracelets) before testing a PSU or working inside your computer.
- Only use the original cables that comes with the tester. The PCIe cable that comes with the tester, is not a standard cable and is custom-made for the tester. Using other cables could lead to short circuit between the voltage rails.
- Make sure all the cables are fully inserted. A loose connection introduces more electrical resistance which can lead to melting of the wires under high load (or even a fire).
- In the case of a dramatic failure of the device under test, you may see smoke coming from the power supply or inside the case or smell a burning. In this case unplug the power supply from the wall. Do not touch the PSU or the PSU Tester in such circumstances.
- Always have someone monitor the testing in case the device under test suddenly fails. Do not start a test and walk away.
- If the PSU Tester is inline (connected to both power supply and motherboard), do not power up using the keypad on the tester. Instead, use the PC power button. This is just a precaution against poor engineered power supplies that do not meet ATX standards and can prevent a short on the PS_ON line.
How do I connect the cables?
- The picture below shows how to connect the tester to a PSU and motherboard using the cables.
- The “SATA In” terminal is located on the side of the device. This connector is a bit tricky as you have to make sure that it has a correct orientation. The ¬ shape of the SATA power cable should face down (See the below picture).
When in inline mode, how should I turn on the PC?
If the PSU Tester is inline (connected to both power supply and motherboard), do not power up using the keypad on the tester. Instead, use the PC power button. This is just a precaution against poor engineered power supplies that do not meet ATX standards and can prevent a short on the PS_ON line.
My PC fails to boot when the PSU tester is inline. What should I do?
First check that all cables are connected properly and fully inserted.
Make sure the “SATA In” cable is plugged and it’s orientation is correct.
Make sure that you turn on the PC using the “standby” button from the computer case and not using the keypad.
If the issue persists, disconnect all the cables to motherboard and force power up by keypad to troubleshoot.
What does the background colour of the PASS/FAIL indicators mean?
The background colour of PASS/FAIL messages can be green, red, light red and yellow. Each colour has a different meaning which are explained below:
PASS means voltage is currently within the approved limits and never been outside the specifications.
PASS means voltage is currently within the approved limits, but an over-voltage was detected previously.
PASS means voltage is currently within the approved limits, but an under-voltage was detected previously.
FAIL means voltage is outside the approved limits.
How can I stress test a PSU?
We suggest running BurnInTest on the motherboard during inline testing to ramp up power usage by the motherboard and place more load on PSU. The recomanded set of test
What is the maximum current/power limit in inline mode?
||+12V2 (12V CPU)
The above ratings correspond to a 1600W PSU. Higher capacity PSUs can also be connected to the tester provided that the current drawn from voltage rails do no exceed the maximum allowed current mentioned in the above table.
What are the timing T1, T2, T3 and T6?
Below you can see the timing diagram for the power good signal as available on the ATX12V specification. “VAC” is the input alternating voltage, i.e., the voltage from the wall. PS_ON# is the “power on” signal (i.e., you pushed the “standby” button from the computer case). “O/P’s” stand for “operating points.” And PWR_OK is the power good signal.
Power-on Time (T1): The power-on time is defined as the time from when PS_ON# is pulled low to when the +12VDC, +5VDC, and +3.3VDC outputs are within the regulation ranges. The power-on time shall be less than 500 ms (T1 < 500 ms).
Rise Time (T2): The output voltages shall rise from ≤10% of nominal to within the regulation ranges within 0.1 ms to 20 ms (0.1 ms ≤ T2 ≤ 20 ms).
PWR_OK delay (T3): PWR_OK is a “power good” signal. It should be asserted high by the power supply to indicate that the +12VDC, +5VDC, and +3.3VDC outputs are above the under-voltage thresholds.
Power-down warning (T6): During power-down, the PWR_OK signal should be asserted low at least 1ms before the +12VDC, +5VDC, and +3.3VDC outputs reach their under-voltage thresholds (T6 ≥ 1ms).
The tester displays N/A for some of the timing values. What does it mean?
“N/A” means some of the timings are not yet measured. Timings are calculated during power-up and power-down process, so if it shows “N/A”, that means one or two power cycles are required to calculate the timings.
Does the device need to be connected to a PC via the USB port?
No. The PSU Tester can be powered up via the 24-Pin connector. However, the voltage reference and the protection circuits in the PSU Tester rely on the 5V input voltage. Therefore, if the +5V rail fails to provide a clean and accurate voltage that could affect the accuracy of the measurements and the functionality of the protection circuits. For that reason, we recommend connecting the USB port to a PC or a wall charger.
Can I measure the efficiency of a PSU using the PSU Tester?
Yes. Measuring efficiency is all about accurately measuring the input power and the output power. The PSU tester can measure the output power. For measuring the input power, a good quality power meter is also required.
Why the tester turns off the PSU after one minute in the standalone mode?
In standalone mode the tester will dissipate 4Watts of power (in order to place a test load on the PSU). This can result in the base of the tester becoming warm to touch after a short period of time. The tester will automatically terminate testing after 1 minute to avoid overheating.
Do I need a device driver to use the PSU Tester?
For Windows 7 and 8, you need a device driver. For Windows 10, the usage of the native inbox driver (Microsoft driver) is recommended.
Is there a Linux driver for the device?
Is there an SDK / API to use the device from my own software?